You cannot calibrate because the test chart is not accepted?
The Konica Minolta Auto Scan Spectrophotometer FD-9 gives the error messge: "Wrong chart detected"?
This error appears if the black line is cut off the test chart. The measurement device needs this black frame for orientation and alignment check.
Around the test chart a colored frame is printed to improve the printer quality. Please crop this for measurment. However, please leave the thin black frame around the test chart: